This bug affects the Standard Edition of ByteLogger for all DMEs. When datalogging with the analog DAQ for the first time, all four channels will display the same voltage as channel #1.
This bug was fixed with the release of ByteLogger v1.9. If you have v1.8 installed, ensure the automatic update feature is enabled and your software should prompt you to download the update when first opened. Please note that the automatic update will not work on Windows XP due to security gaps with that OS. If you purchased an older version of ByteLogger (or have Windows XP) and notice this bug, contact Renovelo for an updated version.
Report a bug with ByteLogger
1 post • Page 1 of 1
- Site Admin
- Posts: 234
- Joined: Thu Aug 08, 2013 7:02 pm